Amendment 1 - Non-invasive sphygmomanometers - Part 2: Clinical investigation of intermittent automated measurement type
Standards Content (sample)
Non-invasive sphygmomanometers —
Clinical investigation of intermittent
automated measurement type
Sphygmomanomètres non invasifs —
Partie 2: Investigation clinique pour type ponctuel à mesurage
---------------------- Page: 1 ----------------------
COPYRIGHT PROTECTED DOCUMENT
© ISO 2020
All rights reserved. Unless otherwise specified, or required in the context of its implementation, no part of this publication may
be reproduced or utilized otherwise in any form or by any means, electronic or mechanical, including photocopying, or posting
on the internet or an intranet, without prior written permission. Permission can be requested from either ISO at the addressbelow or ISO’s member body in the country of the requester.
ISO copyright office
CP 401 • Ch. de Blandonnet 8
CH-1214 Vernier, Geneva
Phone: +41 22 749 01 11
Fax: +41 22 749 09 47
Published in Switzerland
ii © ISO 2020 – All rights reserved
---------------------- Page: 2 ----------------------
ISO (the International Organization for Standardization) and IEC (the International Electrotechnical
Commission) form the specialized system for worldwide standardization. National bodies that
are members of ISO or IEC participate in the development of International Standards through
technical committees established by the respective organization to deal with particular fields of
technical activity. ISO and IEC technical committees collaborate in fields of mutual interest. Other
international organizations, governmental and non-governmental, in liaison with ISO and IEC, alsotake part in the work.
The procedures used to develop this document and those intended for its further maintenance are
described in the ISO/IEC Directives, Part 1. In particular, the different approval criteria needed for
the different types of document should be noted. This document was drafted in accordance with the
editorial rules of the ISO/IEC Directives, Part 2 (see www .iso .org/ directives).
Attention is drawn to the possibility that some of the elements of this document may be the subject
of patent rights. ISO and IEC shall not be held responsible for identifying any or all such patent
rights. Details of any patent rights identified during the development of the document will be in the
Introduction and/or on the ISO list of patent declarations received (see www .iso .org/ patents) or the IEClist of patent declarations received (see http:// patents .iec .ch).
Any trade name used in this document is information given for the convenience of users and does notconstitute an endorsement.
For an explanation of the voluntary nature of standards, the meaning of ISO specific terms and
expressions related to conformity assessment, as well as information about ISO's adherence to the
World Trade Organization (WTO) principles in the Technical Barriers to Trade (TBT) see www .iso .org/iso/ foreword .html.
This document was prepared jointly by Technical Committee ISO/TC 121, Anaesthetic and respiratory
equipment, Subcommittee SC 3, Respiratory devices and related equipment used for patient care,
and Technical Committee IEC/TC 62, Electrical equipment in medical practice, Subcommittee SC D,Electromedical equipment.
A list of all parts in the ISO 81060 series and in the IEC 81060 series can be found on the ISO website.
Any feedback or questions on this document should be directed to the user’s national standards body. Acomplete listing of these bodies can be found at www .iso .org/ members .html.
© ISO 2020 – All rights reserved iii
---------------------- Page: 3 ----------------------
Non-invasive sphygmomanometers —
Clinical investigation of intermittent automated
Add the following after 3.4:
total limb circumference range
range, from the smallest limb circumference to the larges